Main Article Content
This study was aimed to develop an optimized Dy determination method using differential pulse voltammetry (DPV). The Plackett-Burman (PB) experimental design was used to select significant factors that affect the electrical current response, which were further optimized using the response surface method-central composite design (RSM-CCD). The type of electrolyte solution and amplitude modulation were found as two most significant factors, among the nine factors tested, which enhance the current response based on PB design. Further optimization using RSM-CCD shows that the optimum values for the two factors were 0.1046 M and 0.1082 V respectively. When the optimum conditions were applied for Dy determination good recovery and precision were achieved with values of 91.58%, and 99.80%, respectively. The detection limit and quantification limit of the method were 1.4322 mg/L and of 4.7741 mg/L, respectively.
Received 29/9/2019, Accepted 11/5/2020, Published 1/12/2020
This work is licensed under a Creative Commons Attribution 4.0 International License.
Krebs RE. The history and use of our earth's chemical elements: a reference guide. Connecticut: Greenwood Publishing Group; 2006.
Zamani HA, Faridbod F, Ganjali MR. Dysprosium selective potentiometric membrane sensor. Mater. Sci. Eng. C. 2013; 33(2):608-12.
Arslan Z, Oymak T, White J. Triethylamine-assisted Mg(OH)2 coprecipitation/preconcentration for determination of trace metals and rare earth elements in seawater by inductively coupled plasma mass spectrometry (ICP-MS). Anal. Chim. Acta. 2018; 1008:18-28.
Bradley VC, Manard BT, Roach BD, Metzger SC, Rogers KT, Ticknor BW, et al. Rare earth element determination in uranium ore concentrates using online and offline chromatography coupled to ICP-MS. Minerals. 2020; 10(1):55.
Berijani S, Ganjali MR, Sereshti H, Norouzi P. A selective modified nanoporous silica as sorbent for separation and preconcentration of dysprosium in water samples prior to ICP-OES determination. Int J Environ Anal Chem. 2012; 92(3):355-65.
Schramm R. Use of X-ray fluorescence analysis for the determination of rare earth elements. Phys. Sci. Rev. 2016; 1(9):20160061
Jugade R, Jeyaseelan C, Joshi A. Highly sensitive adsorption stripping voltammetric method for determination of trace vanadium. CHEM ANAL-WARSAW. 2005; 50(6):965-72.
Wang J. Analytical Electrochemistry. New Jersey: John Wiley & Sons, Inc.; 2006.
Molina Á, Laborda E, Rogers EI, Martínez-Ortiz F, Serna C, Limon-Petersen JG, et al. Theoretical and experimental study of Differential Pulse Voltammetry at spherical electrodes: Measuring diffusion coefficients and formal potentials. J. Electroanal. Chem. 2009; 634(2):73-81.
Wyantuti S, Pratomo U, Hartati YW, Anggraeni A, Bahti HH. Fast and Simultaneous Detection of Sm, Eu, Gd, Tb and Dy using combination of Voltammetry Method and Multivariate Analysis. Res. J. Chem. Environ. 2018; 22(Special Issue 2):302-6.
Wyantuti S, Pratomo U, Hartati YW, Hendrati D, Bahti HH. A study of green electro-analysis conducted by experimental design method for detection of Samarium as complex with diethylenetriaminepentaacetic acid (DTPA). InAIP Conference Proceedings 2018 Dec 14 (Vol. 2049, No. 1, p. 030010). AIP Publishing LLC.
Miller JN, Miller JC, Miller RD. Statistics and Chemometrics for Analytical Chemistry. 7th ed. Harlow: Pearson Education Limited; Apr 26,2018.
Montgomery DC. Design and Analysis of Experiments. 9th ed. New Jersey: John Wiley & Sons, Inc.; 2017.
Quinlan KR, Lin DK. Run order considerations for Plackett and Burman designs. J. Stat. Plan. Infer. 2015; 165:56-62.
Aksoy DO, Sagol E. Application of central composite design method to coal flotation: Modelling, optimization and verification. Fuel. 2016; 183:609-16.
Mousavi SJ, Parvini M, Ghorbani M. Experimental design data for the zinc ions adsorption based on mesoporous modified chitosan using central composite design method. Carbohydr. Polym. 2018; 188:197-212.
Kola AK, Mekala M, Goli VR. Experimental design data for the biosynthesis of citric acid using central composite design method. Data Brief. 2017; 12:234-41.
Mukherjee A, Banerjee S, Halder G. Parametric optimization of delignification of rice straw through central composite design approach towards application in grafting. J. Adv. Res. 2018; 14:11-23.
Vafaei F, Torkaman R, Moosavian MA, Zaheri P. Optimization of extraction conditions using central composite design for the removal of Co (II) from chloride solution by supported liquid membrane. Chem. Eng. Res. Des. 2018; 133:126-36.
Titah HS, Halmi MIEB, Abdullah SRS, Hasan HA, Idris M, Anuar N. Statistical optimization of the phytoremediation of arsenic by Ludwigia octovalvis-in a pilot reed bed using response surface methodology (RSM) versus an artificial neural network (ANN). Int. J. Phytoremediat. 2018; 20(7):721-9.
Kushkhov HB, Uzdenova AS, Saleh MM, Qahtan AM, Uzdenova LA. The electroreduction of gadolinium and dysprosium ions in equimolar NaCl-KCl melt. Am. J. Anal. Chem. 2013; 4(6A):39-46.
Dejaegher B, Dumarey M, Capron X, Bloomfield M, Vander Heyden Y. Comparison of Plackett–Burman and supersaturated designs in robustness testing. Anal. Chim. Acta. 2007; 595(1-2):59-71.
Ravichandran V, Shalini S, Sundram K, Harish R. Validation of analytical methods–strategies & importance. Int. J. Pharm. Pharm. Sci. 2010; 2(3):18-22.
González AG, Herrador MÁ. A practical guide to analytical method validation, including measurement uncertainty and accuracy profiles. Trend. Anal. Chem. 2007; 26(3):227-38.